- Open Access
Bioequivalance and pharmacokinetic study of febuxostat in human plasma by using LC-MS/MS with liquid liquid extraction method
© Chandu et al.; licensee Springer. 2013
- Received: 22 January 2013
- Accepted: 4 April 2013
- Published: 30 April 2013
A bioequivalence study was proved of generic Febuxostat 80 mg tablets (T) in healthy volunteers.For this purpose, Authors developed a simple, sensitive, selective, rapid, rugged and reproducible liquid chromatography–tandem mass spectrometry method for the quantification of Febuxostat (FB) in human plasma using Febuxostat D7 (FBD7) as an internal standard (IS) was used. Chromatographic separation was performed on Ascentis Express C18 (50x4.6 mm, 3.5 μ) column. Mobile phase composed of 10 mM Ammonium formate: Acetonitrile (20:80 v/v), with 0.8 mL/min flow-rate. Drug and IS were extracted by Liquid- liquid extraction. FB and FBD7 were detected with proton adducts at m/z 317.1→261.1 and 324.2→262.1 in multiple reaction monitoring (MRM) positive mode respectively. The method was validated with the correlation coefficients of (r2) ≥ 0.9850 over a linear concentration range of 1.00-8000.00 ng/mL. This method demonstrated intra and inter-day precision within 2.64 to 3.88 and 2.76 to 8.44% and accuracy within 97.33 to 99.05 and 100.30 to 103.19% for FB. This method is successfully applied in the Bioequivalence study of 9 human volunteers.
- Human plasma
Literature survey reveals that, there are few methods were reported for quantification of Febuxostat by using UV (Bagga et al. 2011;Sheth et al. 2012), Liquid chromatography (LC) (Gunda et al. 2012a;Kumaraswamy Gandla et al. 2012;Cong et al. 2010;Naresh Chandra Reddy & Chandra Sekhar 2012;Nageswara Rao et al. 2012;Gunda et al. 2012b;Muvvala1 et al. 2012;Yamamoto et al. 1995;Menon et al. 2011;Mathrusri Annapurna et al. 2012), UPLC (Sahu et al. 2012), UPLC-MS (Lukram et al. 2012;Zhang et al. 2012). Liquid chromatography -Mass spectrometry (LC) (Kadivar et al. 2011;Ding et al. 2012;Wang et al. 2012). These methods have been reported for the quantitative estimation of Febuxostat in pharmaceutical (Bagga et al. 2011;Sheth et al. 2012;Gunda et al. 2012a;Kumaraswamy Gandla et al. 2012;Cong et al. 2010;Naresh Chandra Reddy & Chandra Sekhar 2012;Nageswara Rao et al. 2012;Gunda et al. 2012b;Muvvala1 et al. 2012;Sahu et al. 2012;Kadivar et al. 2011) and biological fluids (Yamamoto et al. 1995;2011;Menon et al. Lukram et al. 2012;Zhang et al. 2012;Ding et al. 2012;Wang et al. 2012). Among all Quantification of Febuxostat in biological matrices by using LC-MS/MS (Ding et al. 2012;Wang et al. 2012) were reported. Xitao ding et.al (Ding et al. 2012) developed the method in rat plasma using protein precipitation extraction method by using acetonitrile as an precipitating agent. They developed with the linearity range of 10–2000 ng/mL for Febuxostat and used midazolam as internal standard and applied to determinate of Febuxostat in rat plasma for pharmacokinetic study. Wang H et.al. (2012) developed the method in human plasma using protein precipitation extraction method by using acetonitrile as an precipitating agent. They developed the method with linear range of 10.00-5000.00 ng/mL for Febuxostat and used Febuxostat-d7 as an internal standard and performed the pharmacokinetic study in healthy chinese volunteers.
However LC-MS/MS has gained importance for the quantitative estimation of drugs in various biological matrices including plasma, serum, urine, and ocular fluids, due to its high sensitivity, selectivity and reproducibility. The published methods (Ding et al. 2012;Wang et al. 2012) are precipitation methods by using LC-MS. Precipitation method has some disadvantages rather than Liquid-liquid extraction in-terms of extraction efficacy, matrix effect. The aim of the proposed method is extraction of Febuxostat in human plasma by Liquid-liquid extraction method and used internal standard as deuterated compound Febuxostat–d7. The proposed method is most simple, highly sensitive, wide linear and high selective extraction method with good recovery, fully validated as per FDA guidelines and was successfully applied in Bio-equivalence study of 80 mg formulations in healthy human volunteers.
Chemicals and reagents
Febuxostat obtained from AMI Life sciences Pvt.Lt.D Baroda,Gujarat India and Febuxostat D7 Obtained from Euroasian Chemicals Pvt.Ltd, Mumbai , India. HPLC grade methanol, Acetonitrile were purchased from Jt. Baker Mallinckrodt Baker, Inc. Phillipsburg, NJ, USA. Ammonium formate, Formic acid was purchased from S.D fine chemicals Mumbai. Methyl t-butyl ether was purchased from Merck speciality private limited, worli, Mumbai. Ultra pure water from Milli-Q system (Millipore, Bedford, MA, USA) was used through the study. All other chemicals in this study were of analytical grade.Human plasma was obtained from Doctors pathological lab , Hyderabad.
The reference Urolac® immediate release tablet formulation was manufactured by Takeda Pharmaceuticals, America, INC (USA), (batch number R1785, manufacture date Jun 07, 2011, and expiry date Jun 07, 2013). The test formulation was manufactured byAPL Research center. Each immediate release tablet of both formulations contained Febuxostat equivalent to 80 mg. The clinical study was conducted at Clinical and Pharmacological Research Unit, Acron Accunova, Manipal.
HPLC system (1200 Series Agilent Technologies, Germany) connected with triple quadrupole mass spectrometer instrument (API 4000, Toronto, Canada). Data processing was performed with the Analyst 1.4.1 software package (SCIEX). Ionization was performed by Electro spray positive mode with Unit Resolution.
Mass parameters were optimized to get the product ions of m/z: 261.1, m/z: 262.1 from its respective precursor ions of FB [M + H]+ (m/z: 317.1) and FBD7 [M + H]+ (m/z: 324.2) with Source temperature 500°C, Ion Spray voltage 5500 volts, Heater gas, Nebulizer gas 30 psi each, Curtain gas 20 psi, CAD gas 6 psi, (all gas channels with nitrogen) Source flow rate 500 μL/min without split, Entrance potential 10 V, Declustering potential 45 V for analyte and 55 V for internal standard, Collision energy 28 V for both analyte and internal standard, Collision cell exit potential 12 V for analyte and 14 V for internal standard.
Chromatography was performed on Ascentis Express C18 (50x4.6 mm, 3.5 μ) analytical column at 40°C, with 10 mM Ammonium formate: Acetonitrile (20:80 v/v) as mobile phase at a flow rate of 0.8 mL/min. FBD7 was used as an internal standard in terms of chromatography and extractability. The drug and internal standard was eluted at 0.6 ± 0.2 min with 2.5 min total run time.
Preparation of standards and quality control (QC) samples
Standard stock solutions of FB (1000.00 μg/mL) and FBD7 (100.00 μg/mL) was prepared in methanol. The internal standard spiking solution (1000.00 ng/mL) was prepared in 50% acetonitrile from FBD7 standard stock solution (100.00 μg/mL). Standard stock solutions and Internal standard spiking solutions were stored in refrigerator conditions (2–8°C) until analysis. Standard stock solution of FB was added to screened drug-free human plasma to obtain concentration levels of 1.00, 2.00, 20.00, 100.00, 200.00, 400.00, 1000.00, 3000.00, 6000.00, and 8000.00 ng/mL for analytical standards and 1.00, 3.00, 4000.00, 4800.00 ng/mL for Quality control standards and stored in a −30°C freezer until analysis. Respective aqueous standards were prepared in reconstitution solution (10 mM ammonium formate: acetonitrile (20:80) and stored in refrigerator conditions 2–8°C until analysis.
Liquid-liquid extraction was used to isolate drug and IS from human plasma. For this purpose, 100 μL of IS (1000.00 ng/mL) and 100 μL of plasma sample (respective concentration) was added into labeled polypropylene tubes and vortexed briefly. Followed by, 100 μL of 0.1% formic acid, 2.0 mL of extraction solvent (methyl tertiary butyle ether) were added and vortexed for 10 min. Then the samples were centrifuged at 4000 rpm for 5 min at 20°C temperature. Subsequently, the supernatant from each sample was transferred into respective polypropylene tubes. After that, all the samples were kept for evaporation under nitrogen at 40°C. The dried residue was reconstituted with 1000 μL of reconstitution solution and vortexed briefly. Finally, the extracted sample was transferred into auto sampler vials and injected into LC-MS/MS.
Selectivity and specificity
The selectivity of the method was determined by six different human blank plasma samples, which were pretreated and analyzed to test the potential interferences of endogenous compounds co-eluting with analyte and IS. Chromatographic peaks of analyte and IS were identified based on their retention times and MRM responses. The peak area of FB at the respective retention time in blank samples should not be more than 20% of the mean peak area of LOQ of FB. Similarly, the peak area of FBD7 at the respective retention time in blank samples should not be more than 5% of the mean peak area of LOQ of FBD7.
The extraction recovery of FB and FBD7 from human plasma was determined by analyzing quality control samples. Recovery at three concentrations (30.00, 4000.00, and 4800.00 ng/mL) was determined by comparing peak areas obtained from the plasma sample and the standard solution spiked with the blank plasma residue. A recovery of more than 85% was considered adequate to obtain required recovery.
Limit of detection (LOD) and limit of quantification (LOQ)
The limit of detection (LOD) is a parameter that provides the lowest concentration in a sample that can be detected from background noise but not quantitated. LOD was determined using the signal-to-noise ratio (s/n) of 3:1 by comparing test results from samples with known concentrations of analytes with blank samples.
The limit of quantitation (LOQ) is defined as the lowest concentration of analyte that can be determined with acceptable precision and accuracy. The LOQ was found by analyzing a set of mobile phase and plasma standards with a known concentration of FB.
Six lots of blank biological matrices were extracted each in triplicates and post spiked with the aqueous standard at the Low,High QC level, and compared with aqueous standards of same concentration. The overall precision of the matrix factor is expressed as coefficient of variation (CV %) and %CV should be < 15%.
Calibration curve, precission and accuracy
The calibration curve was constructed using values ranging from 1.00 to 8000.00 ng/mL of FB in human plasma. Calibration curve was obtained by linear model with weighted 1/x2 regression analysis. The ratio of FB / FBD7 peak area was plotted against the ratio of FB concentration in ng/mL. Calibration curve standard samples and quality control samples were prepared in replicates (n = 6) for analysis. Precision and Accuracy for the back calculated concentrations of the calibration points, should be within ≤15 and ± 15% of their nominal values. However, for LLOQ, the Precision and Accuracy should be within ≤ 20 and ± 20%.
Stability (Freeze - thaw, Auto sampler, Bench top, Long term) of FB in plasma
Low quality control and high quality control samples (n = 6) were retrieved from a deep freezer after three freeze-thaw cycles according to the clinical protocol. Samples were stored at −30°C in three cycles of 24, 36 and 48 hr. In addition, the long-term stability of FB in quality control samples was also evaluated by analysis after 55 days of storage at −30°C. Autosampler stability was studied following 71.5 hr storage period in the autosampler tray with control concentrations. Bench top stability was studied for 24.5 hr period with control concentrations. Stability samples were processed and extracted along with the freshly spiked calibration curve standards. The Precision and Accuracy for the stability samples must be ≤ 15 and ± 15% respectively of their nominal concentrations.
The study was carried out in accordance with the current revision of the Declaration of Helsinki concerning medical research in humans. Study protocol was approved by IEC (Institutional Ethics committee) as per DCGI (Drug control general of india). Fourteen healthy male subjects, were included in the pilot study. All volunteers gave a written informed consent prior to participation, after they had been informed of the nature and details of the study which they thoroughly understood. Subject screening examinations were performed by a study. Clinician at acron accunova manipal center. All clinical laboratory tests were performed by the ISO 15189 certified laboratories, Department of Pathology, acron accunova manipal Bangalore. The daily results of the clinical laboratory tests including the quality control data were verified by its own independent quality assurance personnel before reporting.
Subject inclusion criteria included Indian male, aged between 18–45 years, no consumption of drugs or food supplements for 4 weeks prior to the study, and no participation in any bioavailability or bioequivalence study at least 30 days prior to the present study.
The exclusion criteria included history of hypersensitivity to Febuxostat and/or related chemical structure and/or any of the components of the product, history or concurrent symptoms of cardiovascular, liver, kidney, gastrointestinal or hematological disorders and/or any disease that might affect the bioavailability of drug, subjects with malignancy, AIDs, allergy, vital sign abnormalities, or clinically significant abnormal values during pre-study screening, smoker (>10 cigarettes/day) or smoker of < 10 cigarettes/day who could not quit at least 7 days before study and throughout study (including washout period), regular alcohol consumption (more than 1 time/week) or alcohol consumption within 7 days prior to the study, coffee consumption within past 7 days, and drug addiction.
The study was conducted as an open label, randomized two-period, two-sequence, single-dose crossover bioequivalence study under fasting condition, and a wash-out period of 7 days. All subjects arrived at the clinical research laboratory, at least 12 h prior to the start of the study. They were housed in an air-conditioned facility and were given a standard dinner, which was finished at least 10 h before dosing in each period of the study. On the day of drug dosing in period 1, volunteers were randomly assigned to one of two treatment sequences (TR (sequence 1) or RT (sequence 2)), as indicated in a pre-printed randomization scheme, which was generated using block randomization with the blocks of size 4 and 6, and the allocation ratio of 1:1. Subjects in sequence 1 received treatment T at the first dosing period and then crossed over to receive treatment R at the second dosing period (after the 7-day washout period). Subjects in sequence 2 received treatments in the order of R and T at the two dosing periods. The subjects were administered the assigned Febuxostat formulation with 240 mL of plain drinking water. After the intake of the study formulations, the oral cavity was checked to ensure completion of the administration process. Subjects were required to refrain from lying down during the first 4 h after dosing.
No meal was permitted until 4 h after dosing. Drinking water was restricted from 1 h before dosing till 2 h after dosing and ad libitum thereafter. Excess water intake (> 100 mL/h) was not permitted. Lunch, snacks, and dinner were served as per the scheduled time. All subjects abstained from any xanthine-containing food or beverages for at least 72 h and alcoholic products for at least 7 days prior to formulation administration and throughout the sampling schedule during each period. They were informed not to take any drug at least 30 days prior to the study, especially phenobarbital, rifampicin or gemfibrozil. Subjects abstained from the use of tobacco- or nicotine-containing products for 7 days prior to dosing and during confinement in the clinical research laboratory. No concomitant medication was permitted during the study period.
Blood samples were collected as the pre-dose (0) hr 5 minutes prior to dosing followed by further samples at 0.167, 0.25, 0.333, 0.5, 0.75, 1, 1.33, 1.667, 2, 2.5, 3, 4, 5, 6, 8, 10, 12, 14, 18, 24, 30 and 36.0 hours. After dosing 2.5 mL blood was collected each time in vaccutainers containing K2EDTA. A total of 44 (23 time points for Reference, 23 time points for Test) time points were collected by using centrifugation 3200 rpm, 10°C, 10 min and stored below −30°C until sample analysis.Test and Reference Febuxostat tablets were administered to same human volunteers under fasting conditions separately with proper washing periods as per protocol (Comparative, Randomized, 2-way crossover) approved by IEC. During the sample collection, all subjects were under medical supervision. Vital signs were examined at scheduled time as described in the protocol.
Throughout the study, subjects were monitored by a clinician, a clinical pharmacist, and 4 nurses. Tolerability was determined by monitoring of vital signs (sitting blood pressure, heart rate, and axillary body temperature), and physical examinations at baseline and at the end of each study period. Subject interviews were also conducted regarding the potential occurrence of adverse events (AEs) at each Febuxostat study period. Serious AEs (SAEs) were considered to be when the subject outcome was death, life threatening, requiring hospitalization, leading to disability, or requiring medical intervention to prevent permanent impairment or damage. All AEs and SAEs were recorded in the source data record and on the case-report form, and their relationship to the study drug was determined by the study physician who was blinded to the randomization schedule.
Pharmacokinetics and statistical analysis
Pharmacokinetics parameters from the human plasma samples were calculated by a non-compartmental statistic model using WinNon-Lin5.0. software (Pharsight, USA). Blood samples were taken for a period of 3 to 5 times the terminal elimination half-life (t1/2) and it was considered as the area under the concentration time curve (AUC) ratio higher than 80% as per FDA guidelines. Plasma FB concentration-time profiles were visually inspected, and Cmax and Tmax values were determined. The AUC0–t was obtained by the trapezoidal method. AUC0–∞ was calculated up to the last measureable concentration and extrapolations were obtained using the last measureable concentration and the terminal elimination rate constant (Ke) it was estimated from the slope of the terminal exponential phase of the plasma of the FB concentration-time curve (by means of the linear regression method). The terminal elimination half-life (t1/2), was then calculated as 0.693/Ke. Regarding AUC0–t, AUC0–∞ and Cmax bioequivalence were assessed by means of analysis of variance (ANOVA) and calculating the standard 90% confidence intervals (90% CIs) of the ratio's test/reference (logarithmically transformed data). The bioequivalence was considered when the ratio of averages of log transformed data was within 80-125% for AUC0–t, AUC0–∞ and Cmax (Guidance for industry 2002;Guidance for industry 2003).
The developed method was validated over a linear concentration range 1.00-8000.00 ng/mL. The validation parameters include Selectivity and Specificity, Limit of Detection (LOD) and Quantification (LOQ), Matrix effect, Precision and Accuracy, Recovery, Stability (Freeze - thaw, Auto sampler, Bench top, Long term) was evaluated under validation section (Guidance for industry 2001).
Selectivity and specificity
Limit of detection (LOD) and quantification (LOQ)
The limit of detection was used to determine the instrument detection levels for FB even at low concentrations. 5 μL of a 0.5 pg/mL solution was injected and estimated LOD was 2.5 fg with S/N values ≥ 3–5.
The limit of quantification for this method was proven as the lowest concentration of the calibration curve which was proven as 1.00 ng/mL.
Six lots of blank biological matrices were extracted each in triplicates and post spiked with the aqueous standard at the mid QC level, and compared with neat standards of same concentration in alternate injections. The overall precision of the matrix factor is 5.67 for Febuxostat. There was no ion- suppression and ion- Enhancement effect observed due to IS and analyte at respective retention time.
Precision and accuracy
Calibration curve details
Spiked plasma concentration (ng/mL)
Concentration measured(ng/mL) mean
CV (%) (n = 5)
Precision and accuracy at four different concentrations
Spiked plasma concen-tration (ng/mL)
(n = 6)
(n = 30)
(mean ± S.d.)
(mean ± S.D.)
0.98 ± 0.04
1.03 ± 0.09
2.92 ± 0.11
3.02 ± 0.17
3957.78 ± 104.45
4021.78 ± 117.78
4754.29 ± 158.36
4814.48 ± 133.07
The recovery following the sample preparation using Liquid-liquid extraction with Methyl tertiary butyle ether was calculated by comparing the peak area of FB in plasma samples with the peak area of solvent samples and was estimated at control levels of FB. The recovery of FB was determined at three different concentrations 3.0, 4000.0 and 4800.0 ng/mL were found as 80.05, 84.02 and 80.67% respectively. The overall average recovery of FB and FBD7 were found to be 81.59 and 89.28% respectively.
Stability (Freeze - thaw, Auto sampler, Bench top, Long term)
Stability of the samples
Spiked plasma concentration (ng/mL)
Room temperature stability
Processed sample stability
Long term stability
Freeze and thaw stability
Cycle 3 (48 h)
CV.a(n = 6) (%)
CVa(n = 6) (%)
CV.a(n = 6) (%)
CV.a(n = 6) (%)
(n = 6)
(n = 6)
(n = 6)
(n = 6)
(mean ± S.D)
(mean ± S.D)
(mean ± S.D)
(mean ± S.D)
3.09 ± 0.16
3.11 ± 0.12
3.03 ± 0.08
3.04 ± 0.16
5034.29 ± 62.49
4800.00 ± 104.74
4800.80 ± 172.50
4923.81 ± 148.20
Application to biological samples
Mean pharmacokinetic parameters of febuxostat in 14 healthy volunteers after oral administration of 80 mg (1x80 mg) test and reference products
AUC0-t (ng · h/ml)
AUC0-∞ (ng · h/ml)
Test/Reference values for Log-transformed pharmacokinetic parameters of febuxostat after administration of 80 mg (1x80 mg) of test and reference products in 14 healthy male volunteers
AUC0 - ∞
The method described in this manuscript has been developed and validated over the concentration range of 1.00 - 8000.00 ng/mL in human plasma. The selectivity, sensitivity, precision and accuracy obtained with this method make it suitable for the purpose of the present study. The simplicity of the method, and using rapid liquid–liquid extraction and sample turnover rate of 2.5 min per sample, make it an attractive procedure in high-throughput bioanalysis of Febuxostat. The validated method was successfully applied and demonstrated the bioequivalence of the 80-mg immediate release tablet formulation manufactured by APL Research center., India and the reference product Urolac® manufactured by Takeda Pharmaceuticals, America,INC(USA) by oral administration in 14 healthy human volunteers, and then can be concluded that the two formulations can be used interchangeably.
Authors are greatful to Acron accunova for providing opportunity to participating in study monitoring and research activities. Authors wish to thank the support received from IICT (Indian institute of chemical technology) Hyderabad India for providing Literature survey.
- Bagga P, Salman M, Siddiqui HH, Abdul M, Ansari TM, Singh K: A simple UV spectrophotometric method for the determination of febuxostat in bulk and pharmaceutical formulations. IJPSR 2011, 2(10):2655-2659.Google Scholar
- Becker MA, Schumacher HR Jr, Wortmann RL, MacDonald PA, Palo WA, Eustace D, Vernillet L, Joseph-Ridge N: Febuxostat, a novel nonpurine selective inhibitor of xanthine oxidase: a twenty-eight-day, multicenter, phase II, randomized, double-blind, placebo-controlled, dose–response clinical trial examining safety and efficacy in patients with gout. Arthritis Rheum 2005, 52(3):916-923. 10.1002/art.20935View ArticleGoogle Scholar
- Cong Z, Shao-jie W, Rong-li MA, Ping M, Tian-hong Z: Determination of content of Febuxostat and its related substances by HPLC. J Shenyang Pharm Univ 2010, 27(08):648-651.Google Scholar
- Ding X, Zhang Q, Wang Z, Lin G: Development and validation of liquid chromatography-mass spectrometry method for determination of febuxostat in Rat plasma and its application Lat. Am J Pharm 2012, 31(2):321-325.Google Scholar
- Grabowski BA, Khosravan R, Vernillet L, Mulford DJ: Metabolism and excretion of Febuxostat, a novel nonpurine selective inhibitor of xanthine oxidase, in healthy male subjects. J Clin Pharmacol 2011, 51(2):189-201. 10.1177/0091270010365549View ArticleGoogle Scholar
- Guidance for industry: Bioanalytical method validation. U.S. Department of Health and Human Services, Food and Drug Administration, Center for Drug Evaluation and Research (CDER), Center for Biologics Evaluation and Research (CBER); 2001.Google Scholar
- Guidance for industry: Food- effect bio availability and Fed Bio equivalence studies. U.S Department of Health and Human services Food and Drug Administration Centre for Drug Evaluation and research (CDER); 2002.Google Scholar
- Guidance for industry: Bio availability and Fed Bio equivalence Studies for Orally Administered Drug Products-General considerations. U.S. Department of Health and Human services Food and Drug Administration Centre for Drug Evaluation and research (CDER); 2003.Google Scholar
- Gunda A, Aravindsai N, Karnaker Reddy T, Anand Kumar A: Estimation of Febuxostat drug present in formulation by RP-HPLC. J Pharm Res 2012, 5(2):1224-1227.Google Scholar
- Hoshide S, Takahashi Y, Ishikawa T, Kubo J, Tsuchimoto M, Komoriya K, Ohno I, Hosoya T: PK/PD and safety of a single dose of TMX-67 (Febuxostat) in subjects with mild and moderate renal impairment. Nucleosides Nucleotides Nucleic Acids 2004, 23(8–9):1117-1118.View ArticleGoogle Scholar
- Hussar DA, Bilbow C: New drugs: Febuxostat, lacosamide, and rufinamide. J Am Pharm Assoc 2009, 49(3):460-463. 10.1331/JAPhA.2009.09516View ArticleGoogle Scholar
- Kadivar MH, Sinha PK, Kushwah D, Jana P, Sharma H, Bapodra A: Study of impurity carryover and impurity profile in Febuxostat drug substance by LC-MS/MS technique. J Pharm Biomed Anal 2011, 56(4):749-757. 10.1016/j.jpba.2011.07.039View ArticleGoogle Scholar
- Kamatani N, Fujimori S, Hada T, Hosoya T, Kohri K, Nakamura T, Ueda T, Yamamoto T, Yamanaka H, Matsuzawa Y: Multicenter, open-label study of long-term administration of Febuxostat (TMX-67) in Japanese patients with hyperuricemia including gout. J Clin Rheumatol 2011, 17(4 Suppl 2):S50-S56.View ArticleGoogle Scholar
- Khosravan R, Grabowski B, Wu JT, Joseph-Ridge N, Vernillet L: Effect of food or antacid on pharmacokinetics and pharmacodynamics of Febuxostat in healthy subjects. Br J Clin Pharmacol 2008, 65(3):355-363. 10.1111/j.1365-2125.2007.03016.xView ArticleGoogle Scholar
- Kumaraswamy Gandla JMR, Kumar DVRNB, Gajjela V, Spandana R: A validated RP-HPLC method for simultaneous estimation of febuxostat and ketorolac tromethamine in pharmaceutical formulations. J Drug Deliv Ther 2012, 2(3):173-176.Google Scholar
- Lukram O, Parmar S, Hande A: Determination of Febuxostat in human plasma using ultra-performance liquid chromatography tandem mass spectrometry. Drug Test Anal 2012, . [Epub ahead of print] 10.1002/dta.420Google Scholar
- Mathrusri Annapurna M, Sai Pavan Kumar B, Venkatesh B, Chitaranjan M: Development and validation of a stability-indicating RP-HPLC method for the determination of febuxostat (a xanthine oxidase inhibitor). J Chromatogr Sci 2012, 1-8. 10.1093/chromsci/bms192Google Scholar
- Menon S, Kandari K, Kadam P: Bioequivalence and pharmacokinetics evaluation of Two formulation of febuxostat 80 Mg tablets in healthy Indian adult subject. Int J Curr Pharm Rev Res 2011, 2(2):92-100.Google Scholar
- Muvvala1 SS, Nadh Ratnakaram V, Rao Nadendla R: A validated RP-HPLC method for the estimation of febuxostat in bulk drugs. Int J Pharm Tech Res 2012, 4(4):1358-1366.Google Scholar
- Nageswara Rao K, Ganapaty S, Lakshmana Rao A: Development and validation of RP-HPLC method for estimation of febuxostat in bulk and tablet dosage form. Int J Res Pharm Chem 2012, 2(4): .Google Scholar
- Naresh Chandra Reddy M, Chandra Sekhar KB: Estimation of related substances of febuxostat in bulk & 40/80/120 mg tablets by Rp-hplc. IJPBCS 2012, 1(2):01-10.Google Scholar
- Sahu K, Shaharyar M, Siddiqui AA: Establishment of inherent stability of Febuxostat and development of a validated stability-indicating method by UPLC according to ICH requirement. Med Chem Res 2012. 10.1007/s00044-012-0160-5Google Scholar
- Sheth M, Joshi S, Patel M: Development and application of difference spectrophotometric method for the determination of febuxostat in tablets. IJPSR 2012, 3(06): .Google Scholar
- Tayar JH, Lopez-Olivo MA, Suarez-Almazor ME: Febuxostat for treating chronic gout. Cochrane Database Syst Rev 2012. 10.1002/14651858.CD008653.pub2Google Scholar
- Tomillero A, Moral MA: Gateways to clinical trials. Methods Find Exp Clin Pharmacol 2009, 31(3):183-226.Google Scholar
- Wang H, Deng P, Chen X, Guo L, Zhong D: Development and validation of a liquid chromatography-tandem mass spectrometry method for the determination of Febuxostat in human plasma. Biomed Chromatogr 2012, . 10.1002/bmc.2744Google Scholar
- Yamamoto T, Moriwaki Y, Takahashi S, Nasako Y, Yamakita J, Hiroishi K, Higashino K: Determination of plasma purine nucleoside phosphorylase activity by high-performance liquid chromatography. Anal Biochem 1995, 227(1):135-139. 10.1006/abio.1995.1262View ArticleGoogle Scholar
- Zhang T, Sun Y, Zhang P, Gao J, Wang S, He Z: Ultra-performance liquid chromatography-tandem mass spectrometry method for the determination of Febuxostat in dog plasma and its application to a pharmacokinetic study. Biomed Chromatogr 2012, . 10.1002/bmc.2756Google Scholar
- Zhu X, Wang Y, Lu T: 2-[3-Cyano-4-(2-methyl-prop-oxy)phen-yl]-4-methyl-thia-zole-5-carboxylic acid pyridine solvate. Acta Crystallogr Sect E Struct Rep Online 2009, 65(Pt 11):02603.View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.