Single-breath-hold whole-heart coronary MRA in healthy volunteers at 3.0-T MRI
© Iyama et al.; licensee Springer. 2014
Received: 13 June 2014
Accepted: 5 November 2014
Published: 11 November 2014
The purpose of this study was to investigate the feasibility of single-breath-hold whole-heart MRA with a 3-T system. Ten healthy male volunteers underwent single-breath-hold whole-heart coronary MRA at 3 T. We assessed acquisition time, scores of image quality of coronary artery (RCA: proximal, middle and distal, LAD: main, proximal, middle and distal, LCX: proximal and distal) and the visualized vessel length of RCA, LAD and LCX.
Mean acquisition time was 37.7 ± 5.2 sec. Coronary branch was successfully depicted in 67/80 branches (84%) in the 10 healthy volunteers with diagnostic image quality. And, the average visible RCA, LAD and LCX vessel length were 83.4 ± 22 mm and 59.6 ± 24 mm.
3-T MRI with single-breath-hold 3D whole-heart coronary MRA can yield adequate image quality. Further study is needed to evaluate the clinical benefit of this technique.
Whole-heart coronary MR angiography (MRA) is generally carried out during free breathing with a respiratory gating method using navigator echo techniques, which track the motion of the right hemi-diaphragmatic dome. The advantage of free breathing technic is unnecessary to stop breathing. However, the major drawback of this free breathing technique is the relatively long acquisition time, ranging from 10 to 20 minutes (Sakuma et al. 2005). The technic of single-breath hold technic could shorten the total scan time because it can reduce influence of respiratory motion. There were some reports about whole-heart coronary MRA with 1.5 T MRI during single breath hold (Makowski et al. 2012; Okada et al. 2011; Nassenstein et al. 2008). In general, increased SNR by 3-T MRI enables us to increase image quality and exam speed compared to 1.5-T MRI; however, high-field systems still pose challenges in terms of their specific absorption rate (SAR) and radiofrequency (RF) excitation uniformity, especially in cardiac MRI (Nezafat et al. 2006).
Recently introduced 3-T systems encompass two major innovative technologies for coronary MRA. The Direct Digital RF receiver technology digitizes the MR signal at the patient, and the fiber-optic connection from the coil to the image reconstructor enables lossless broadband data transmission. Previous reports suggested that this technique improved the dynamic range of the RF receiver and resulted in an improved signal-to-noise ratio (SNR) (Ruipeng et al. 2009). Dual-source radiofrequency transmission with patient-adaptive local radiofrequency shimming enables uniform RF shimming for cardiac MRI (Mueller et al. 2012). This technique reduces dielectric shading, improves B 1 homogeneity, and increases image contrast by T2 preparation prepulse (T2prep) with high-power refocusing pulses. These new technologies in 3-T MRI might increase the SNR at cardiac MRI, and enable us to shorten the acquisition time of whole-heart coronary MRA with adequate image quality. However, to our knowledge, there is no published protocol for single-breath-hold whole-heart MRA with recent 3-T systems.
The purpose of this study was to investigate the feasibility of single-breath-hold whole-heart MRA with a 3-T system on healthy volunteers.
Volunteer’s objective data
Total scan time (sec)
35.6 ± 11.0
59.8 ± 9.6
37.7 ± 5.2
170.6 ± 3.3
66.4 ± 6.3
87.3 ± 22.5
61.5 ± 21.7
114.9 ± 29.3
MR angiography acquisition
All ten subjects were imaged by 3-T MRI (Ingenia, Philips Medical Systems) using a 16-element phased-array Direct Digital RF receiver coil and vector electrocardiographic (VCG) gating (Fischer et al. 1999). A multi-slice gradient echo (TR = 2.6 ms; TE = 1.27 ms; α = 20°) scout scan was acquired in 3 orthogonal orientations for localization of the volume for whole-heart imaging. After the 3D scout scan, an axial ECG-triggered, segmented steady-state free precession (SSFP) cine image series (TR = 2.6 ms, TE = 1.28 ms, α = 45°, and temporal resolution of 10 ms) at the level of the proximal-to-mid right coronary artery (RCA) was also obtained during a single breath hold. This was carried out for visual determination of the most quiescent period in the cardiac cycle, which was subsequently used to set the trigger delay and the shot duration. In addition, an ECG-triggered segmented 3 D SSFP sequence using the proposed undersampling scheme was implemented and in-vivo measurements were performed in expiration.
ACQ voxel size
2.0 × 2.0 × 2.0 mm
Reconstructed voxel size
1.0 × 1.0 × 1.0 mm
2 mm (1 mm reconstruction)
Fast imaging mode
Y factor: 0.625, Z factor: 0.9
TE: 70 ms, 4 RF pulses
Phase direction 2, Slice direction 1.5
To evaluate the image quality, we performed qualitative image analysis of axial images and curved MPR images on a PACS viewer (Synapse, Fuji Film Medicals). Two board-certified radiologists with 8 and 5 years of experience with cardiac MRI independently graded overall image quality according to a segmentation scheme recommended by the American College of Cardiology and the American Heart Association (ACC/AHA) (Scanlon et al. 1999). The RCA was subdivided into three segments (proximal, middle and distal), the LAD into four segments (main, proximal, and middle) and the LCX into two segments (proximal and distal). We defined these segment as follows; (#1 in AHA: RCA proximal segment, #2 in AHA: RCA middle segment, #3 in AHA: RCA distal segment, #5 in AHA: LAD main segment, #6 in AHA: LAD proximal segment, #7 in AHA: LAD middle segment, #11 in AHA: LCX proximal segment and #13 in AHA: LCX distal segment). We used a 5-point subjective scale for qualitative image analysis: 4, excellent (the vessel was well depicted with sharply defined borders); 3, good (the vessel was adequately visualized, with confidence in the diagnosis, only mildly blurred borders); 2, fair (coronary vessel was visible, but confidence in the diagnosis was low due to moderately blurred borders); 1, poor (coronary vessel was barely seen or was obscured by noise); and 0, not visualized (Sakuma et al. 2005; Wu et al. 2007). The visualized vessel length of RCA, LAD and LCX were also measured. Interobserver disagreements were resolved by consensus.
Single-breath-hold whole-heart coronary MRA was technically successful in all 10 volunteers. Total scan times varied from 31 sec to 45 sec because we changed the shot duration for each volunteer.
Well-depicted coronary branch (score: 4 or 3)
RCA proximal segment (#1)
3.5 ± 0.7
3.7 ± 0.7
3.6 ± 0.7
RCA middle segment (#2)
3.5 ± 0.7
3.7 ± 0.7
3.6 ± 0.7
RCA distal segment (#3)
2.5 ± 0.7
3.7 ± 0.7
3.1 ± 1.0
LAD main segment (#5)
2.7 ± 0.5
3.6 ± 0.7
3.3 ± 0.7
LAD proximal segment (#6)
2.4 ± 0.5
3.5 ± 0.7
3.1 ± 0.7
LAD middle segment (#7)
2.7 ± 0.5
3.3 ± 0.9
3.0 ± 0.7
LCX proximal segment (#11)
2.8 ± 0.6
3.2 ± 0.9
3.2 ± 0.8
LCX distal segment (#13)
2.7 ± 0.5
3.3 ± 0.9
3.1 ± 0.7
The purpose of this study was to investigate the feasibility of single-breath-hold whole-heart MRA with a 3-T system. We assessed acquisition time, scores of image quality of coronary artery of ten volunteers. Mean acquisition time was 37.7 ± 5.2 sec. The average visible RCA, LAD and LCX vessel length were 83.4 ± 22 mm and 59.6 ± 24 mm.
To our knowledge, this is the first report on the clinical feasibility of single-breath-hold 3D whole-heart coronary MRA in 3-T MRI.
Although single-breath-hold 3D whole-heart coronary MRA has been performed at 1.5-T MRI using the SSFP technique, the increased B1 field inhomogeneity and SAR limit the consistency of SSFP in coronary images in 3.0-T MRI (Nezafat et al. 2006; Stuber et al. 2002). There are a few reports about 1.5 T single-breath-hold coronary MRA (Makowski et al. 2012; Lim et al. 2013). Because, this technic has disadvantage of breathing hold for a long time, therefore, patients with a respiratory disease have difficulty in receiving examination. However, 3 T MRI can shorten scan time, and patients may not need a long breath-hold. Therefore, TFE sequence that has better tolerance to field inhomogeneity than SSFP has been used for coronary MRA at 3.0 T. However, a major drawback of coronary MRA with TFE is that the SNR of the coronary arteries and the blood-myocardial contrast are not as high as those of SSFP sequence (Maintz et al. 2004). And, there were few reports about single-breath-hold 3D whole-heart coronary MRA with our study suggested that a recent 3-T MRI might offer adequate SNR in spite of having the short acquisition time of whole-heart coronary MRA. We did not know the reason the result of our study was better than that of previous report about single-breath-hold 3D whole-heart coronary MRA (Nezafat et al. 2006; Stuber et al. 2002). There are many factors to improve image quality ( for example, recent 3-T MRI system, such as increasing the number of channel coils, the Direct Digital RF receiver technology and the dual-source radiofrequency transmission). We think that the increased refocusing pulses (T2prep) mainly overcome the decreased signal with the TFE sequence in free breathing coronary MRI. In general, the image contrast upon using TFE MRI is dependent on the number of prepulses. Botnar et al. state that the combined approach of free-breathing navigator-gated and slice-tracked 3D coronary MRA together with a T2prep and a shorter acquisition window resulted in an improved CNR between coronary blood and myocardium and thereby allowed for better definition of the coronary vessels (Maintz et al. 2004). And, T2 prep technic may be useful in breath-hold coronary MRI.
Actually, our study shows that the average visible RCA vessel length of 114.9 ± 29.3 mm compared favorably with earlier reported navigator-gated bSSFP (80 ± 40 mm) and gradient echo sequences (95 ± 22 mm) in RCA at 3.0 T (Kaul et al. 2004). Therefore, a large value of T2prep could improve image quality. Sahar et al. reported that T2 prep technic increased image quality compared to non T2 prep technic (Soleimanifard et al. 2013). Mueller et al. state that dual-source RF transmission with RF shimming results in an optimized SAR distribution, thereby reducing local SAR peaks (Mueller et al. 2012). As such, we can decrease SAR by using multitransmit technology.
Our study had a number of limitations. First, we only evaluated 10 healthy volunteers. Future studies are needed to evaluate more patients with suspected coronary heart disease. Second, our evaluation only involved a protocol for single-breath-hold whole-heart MRA with the recent 3-T systems. Future studies are needed to compare the protocol for a single breath hold and the protocol for free breathing with a respiratory gating method. Third, total scan time in our study is comparatively long (from 31 sec to 45 sec). However, increasing reduction factor in parallel imaging be able to shorten total scan time in future studies. Forth, in our study, the image quality of the distal segments of the LCX and LAD were significantly poorer than the image quality of the RCA. When the distal segments were excluded, the image quality between the coronary arteries was not significant different. We believe that the poor quality of the LCX and LAD was caused by the small diameter of its distal segment. The result of the previous report was the same kind (Kim et al. 2006).
In conclusion, 3-T MRI with single-breath-hold 3D whole-heart coronary MRA can yield adequate image quality. Further study is needed to evaluate the clinical benefit of this technique.
- Fischer SE, Wickline SA, Lorenz CH: Novel real-time R-wave detection algorithm based on the vectorcardiogram for accurate gated magnetic resonance acquisitions. Magn Reson Med 1999, 42(2):361-370. doi:10.1002/(SICI)1522-2594 10.1002/(SICI)1522-2594(199908)42:2<361::AID-MRM18>3.0.CO;2-9View ArticleGoogle Scholar
- Kaul MG, Stork A, Bansmann PM, Nolte-Ernsting C, Lund GK, Weber C, Adam G: Evaluation of balanced steady-state free precession (TrueFISP) and K-space segmented gradient echo sequences for 3D coronary MR angiography with navigator gating at 3 Tesla. Röfo 2004, 176(11):1560-1565. doi:10.1055/s-2004-813629Google Scholar
- Kim YJ, Seo JS, Choi BW, Choe KO, Jang Y, Ko YG: Feasibility and diagnostic accuracy of whole heart coronary MR angiography using free-breathing 3D balanced turbo-field-echo with SENSE and the half-fourier acquisition technique. Korean J Radiol 2006, 7(4):235-242. 10.3348/kjr.2006.7.4.235View ArticleGoogle Scholar
- Lim RP, Winchester PA, Bruno MT, Xu J, Storey P, McGorty K, Sodickson DK, Srichai MB: Highly accelerated single breath-hold noncontrast thoracic MRA: evaluation in a clinical population. Invest Radiol 2013, 48(3):145-151. doi:10.1097/RLI.0b013e31827994c2View ArticleGoogle Scholar
- Maintz D, Aepfelbacher FC, Kissinger KV, Botnar RM, Danias PG, Heindel W, Manning WJ, Stuber M: Coronary MR angiography: comparison of quantitative and qualitative data from four techniques. AJR Am J Roentgenol 2004, 182(2):515-521. doi:10.2214/ajr.182.2.1820515View ArticleGoogle Scholar
- Makowski MR, Wiethoff AJ, Jansen CH, Uribe S, Parish V, Schuster A, Botnar RM, Bell A, Kiesewetter C, Razavi R, Schaeffter T, Greil GF: Single breath-hold assessment of cardiac function using an accelerated 3D single breath-hold acquisition technique--comparison of an intravascular and extravascular contrast agent. J Cardiovasc Magn Reson 2012, 14: 53. doi:10.1186/1532-429x-14-53View ArticleGoogle Scholar
- Mueller A, Kouwenhoven M, Naehle CP, Gieseke J, Strach K, Willinek WA, Schild HH, Thomas D: Dual-source radiofrequency transmission with patient-adaptive local radiofrequency shimming for 3.0-T cardiac MR imaging: initial experience. Radiology 2012, s263(1):77-85. doi:10.1148/radiol.11110347View ArticleGoogle Scholar
- Nassenstein K, Waltering KU, Kelle S, Schlosser T, Breuckmann F, Maderwald S, Hunold P, Nagel E, Barkhausen J: Magnetic resonance coronary angiography with Vasovist: in-vivo T1 estimation to improve image quality of navigator and breath-hold techniques. Eur Radiol 2008, 18(1):103-109. doi:10.1007/s00330-007-0720-0View ArticleGoogle Scholar
- Nezafat R, Stuber M, Ouwerkerk R, Gharib AM, Desai MY, Pettigrew RI: B1-insensitive T2 preparation for improved coronary magnetic resonance angiography at 3 T. Magn Reson Med 2006, 55(4):858-864. doi:10.1002/mrm.20835View ArticleGoogle Scholar
- Okada T, Kanao S, Kuhara S, Ninomiya A, Fujimoto K, Kido A, Togashi K: Whole-heart coronary MR angiography under a single breath-hold: a comparative study with respiratory-gated acquisition using a multi-element phased-array coil. Clin Radiol 2011, 66(11):1060-1063. doi:10.1016/j.crad.2011.06.004View ArticleGoogle Scholar
- Ruipeng N, Yidong D, Guang Y, Gengying L: A digital receiver with fast frequency- and gain-switching capabilities for MRI systems. MAGMA 2009, 22(6):333-342. doi:10.1007/s10334-009-0182-2View ArticleGoogle Scholar
- Sakuma H, Ichikawa Y, Suzawa N, Hirano T, Makino K, Koyama N, Van Cauteren M, Takeda K: Assessment of coronary arteries with total study time of less than 30 minutes by using whole-heart coronary MR angiography. Radiology 2005, 237(1):316-321. doi:10.1148/radiol.2371040830View ArticleGoogle Scholar
- Scanlon PJ, Faxon DP, Audet AM, Carabello B, Dehmer GJ, Eagle KA, Legako RD, Leon DF, Murray JA, Nissen SE, Pepine CJ, Watson RM, Ritchie JL, Gibbons RJ, Cheitlin MD, Gardner TJ, Garson A Jr, Russell RO Jr, Ryan TJ, Smith SC Jr: ACC/AHA guidelines for coronary angiography. A report of the American College of Cardiology/American Heart Association Task Force on practice guidelines (Committee on Coronary Angiography). Developed in collaboration with the Society for Cardiac Angiography and Interventions. J Am Coll Cardiol 1999, 33(6):1756-1824. 10.1016/S0735-1097(99)00126-6View ArticleGoogle Scholar
- Soleimanifard S, Schar M, Hays AG, Prince JL, Weiss RG, Stuber M: Spatially selective implementation of the adiabatic T2Prep sequence for magnetic resonance angiography of the coronary arteries. Magn Reson Med 2013, 70(1):97-105. doi:10.1002/mrm.24437View ArticleGoogle Scholar
- Stuber M, Botnar RM, Fischer SE, Lamerichs R, Smink J, Harvey P, Manning WJ: Preliminary report on in vivo coronary MRA at 3 Tesla in humans. Magn Reson Med 2002, 48(3):425-429. doi:10.1002/mrm.10240View ArticleGoogle Scholar
- Wu YW, Tadamura E, Yamamuro M, Kanao S, Nakayama K, Togashi K: Evaluation of three-dimensional navigator-gated whole heart MR coronary angiography: the importance of systolic imaging in subjects with high heart rates. Eur J Radiol 2007, 61(1):91-96. 10.1016/j.ejrad.2006.08.013View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited.